|Heating Power::||1800W||Power Supply::||Ac 220V±10%,50HZ±5%;|
|Ambient Temperature::||5℃-40℃;||Environmental Humidity::||< 85%;|
chemical composition analysis equipment,
chemical analysis machine
SH112E low-temperature petroleum kinematicity viscometer this instrument conforms to the national standard GB/T265 "petroleum products kinematicity viscometry method and dynamic viscometry calculation method" standard and NB/SH/T0837 mineral insulation oil low-temperature kinematicity viscometry measurement requirements.
Can install 2 viscosity at the same time and carry out the experiment.
The main features are as follows:
1. The thermostatic bath is a low-temperature double-layer dewar bottle, with even temperature distribution and good temperature control effect.
2, this instrument USES the precision digital temperature controller to control the temperature in the bathtub, the execution component USES the solid state relay SSR, has the temperature control accuracy accuracy, no contact, no noise, no spark, fast action, long life and other characteristics.
3, the capillary card adopts three-point vertical, flexible and convenient operation, reliable clamping.
4, standard with micro printer automatic measurement automatic printing
5. It can also include a laser printer interface, which can be connected to an HP1108 laser printer (optional).
Main technical indicators:
The temperature is measured with precision platinum resistance PT100. The temperature is set arbitrarily at 0 ~ 100℃. The main and auxiliary heaters are automatically switched.
Automatic calculation of viscosity value.
Temperature control point: -40℃-60℃ arbitrarily set temperature control precision: ±0.1℃;
Display mode: digital display;
Stopwatch timing: 0.05 seconds --99 minutes 99.99 seconds;
Heating power: 1800W;
Power supply: ac 220V±10%,50HZ±5%;
Ambient temperature: 5℃-40℃;
Environmental humidity: < 85%;
External dimension: 190mm×380mm×280mm;
Customer's own supplies: water or alcohol.
Strictly according to the national standard, the viscosity tube test needs the following reagent cleaning: solvent oil or petroleum ether.