What is the water content of the new oil standard?
Many years of operation experience shows that in many cases, the acid value of the oil in operation is increased because of the hydrolysis of the oil, so the new oil standard is <.
The moisture standard of running oil is 1000 mg/L.
Moisture: Test according to GB/T 7600 method.
Water will not only lead to hydrolysis degradation of phosphate ester fuel oil, acid value increase, resulting in the corrosion of system components, but also affect the lubrication characteristics of the oil.
If the moisture content of the running phosphoric ester fuel exceeds the standard, the reasons should be found out quickly and effective treatment measures should be taken.
SH103 micro moisture meter (shandong shengtai instrument co., LTD.) adopts according to the principle of Karl fischer coulomb titration, using microcomputer automatic control, compared with similar products at home and abroad, it has the advantages of high sensitivity, fast electrolysis speed, short balance time, accurate blank deduction, accurate and reliable analysis results.
It is a high quality analytical instrument for petroleum, chemical, electric power, environmental protection, medicine, scientific research and other departments.
The instrument adopts 32-bit embedded microprocessor as the main control core, embedded in the mini-operating system;
The instrument comes with constant current detection, double circuit balance titration function;
The electrolytic tank adopts waterproof and sealed design;
Support multiple calculation formulas, in line with the standard GBT11133 and ASTMD 0304.
Carl, -- Hugh coulomb's law, to determination of trace moisture of different materials, in line with the GB/T11133, this trace moisture meter and the success of the application of this method, and used the micro computer control, its fast analysis speed, high precision, Chinese LCD color screen display, automatic printing, double CPU design, blank current deducted automatically and instrument fault self-diagnosis function,
Menu selection and other functions to achieve better operation and use, with a simple operation, automatic analysis instrument.
It is an instrument for oil moisture analysis
Display mode: color touch screen display
Titration method: microcomputer controlled electric titration
Output mode: automatic calculation, can output ug, PPM and %
Measurement range: 0.00001% to 100% as a percentage, 0.1 PPM to 1 million PPM as per PPM
Separation rate: 0.1ug
Electrolysis control: automatic electrolysis current tracking control (up to 400mA)
Titration speed: 2.5mg/min(Max)
Sensitivity: 0.1ug-- 100g not more than 2%, converted to 0.3% above 1mg (excluding injection error)
Power source: AC 220V± 10% 50Hz± 10%
Power: 60 va
Ambient temperature: 5-40~C
Operating ambient temperature: ≤ 90%
Required consumables: special Karl Fischer reagents, the instrument itself with a set of reagents, after use can be purchased or equipped by themselves.
DL/T 915-05, GB 5832-1986
Meet the following standards:
1, GB/T7600 "in operation transformer oil moisture content determination method (coulomb method)"