Reasons for lowering the flash point of lubricating oil
The flash point of the lubricating oil is the lowest temperature at which the oil surface can flicker under the specified test conditions.
When the volatile oil and gas on the surface of the oil mixed with the oxygen in the air, if the flash point is low, it is easy to flash, so the main problem of the drop of the flash point of the lubricating oil is likely to cause combustion and even fire problems, if the flash will damage the seal and cause leakage.
The higher the flash point of the lubricating oil, the better, but the higher the flash point does not mean that the lubricating oil has excellent high temperature performance.
Cracking of base oil is one of the reasons for the decrease of flash point of lubricating oil. When oil or equipment is near high temperature heat source, such as high temperature steam, high temperature furnace, etc., if the oil temperature of lubricating oil is higher than 550℃, cracking may occur, resulting in the decrease of flash point of lubricating oil.
SH106B automatic open flash point tester (developed and produced by Shandong Shengtai Instrument Co., Ltd.) is designed and manufactured according to the requirements stipulated in the People's Republic of China standard GB/T 3536 "Determination of flash point and ignition point of petroleum products (Cleveland open cup method)".
Instrument according to the standard method of heating, automatic rise and fall, automatic ventilation, automatic ignition, automatic display, automatic locking flash point value, automatic printing results.
After the test can be automatically cooled, to achieve the full automation of the work process.
It has the advantages of accurate measurement, good repeatability, stable performance and simple operation. It also supports the determination of GB/T1671-2008 plasticizer flash point
The instrument features
1. The instrument adopts color LCD large screen display, full Chinese man-machine dialogue interface, no identification keyboard, for the parameters such as pre-value temperature, sample label, atmospheric pressure, test date, with prompt menu, oriented input function.
2. Simulation tracking displays the function curve of temperature rise and test time. It has the function of Chinese wrong operation software to prompt modification, and the function of test date, test time and other parameters to prompt.
3. Automatically correct the influence of atmospheric pressure on the test and calculate the correction value.
4. Differential detection, automatic correction of system deviation.
5. Scanning, detection and printing data are automatically completed, and the test arm automatically rises and falls.
6. Electronic ignition and forced air cooling
7, without automatic ignition of air source, improve the safety of the test
Main technical indexes and parameters
1, working power supply: AC220V±10%, 50Hz
2, temperature measurement: range: room temperature ~ 400℃;
Repeatability: flash point ≤150℃ error ±2℃ flash point ≥150℃ error ±3℃
3. Heating speed: in line with GB/T3536 standard;
4, ignition mode: electronic ignition, gas flame 3㎜~4㎜;
5, ambient temperature: (10~40)℃;
6, relative humidity: ≤80%;
Contact Person: Miss.