|Name:||Degassing Oscillator||Display::||Large-screen LCD|
|Temperature Range::||Room Temperature ~100℃||Temperature Control Accuracy::||Room Temperature ~50℃±0.2℃|
|Oscillation Frequency::||275±3 Times/min;||Oscillation Amplitude::||35mm;|
lube oil analysis equipment,
oil moisture meter
SH121 degassing oscillator
GB/T17625 DL429.4 standard, used for the content determination of dissolved gas components in oil (gas chromatography);
The oscillating degassing of water-soluble acid in oil (colorimetric method) can also be used in other physical and chemical experiments.
1. Determination of water-soluble acid in transformer oil and turbine oil in operation gb7598-87 (colorimetric method);
2. Sd304-89 method for determination of dissolved gas components in insulating oil (gas chromatography);
3. Gb259-88 method for determination of water-soluble acids and bases in petroleum products.
The main features
Blue LCD display, Chinese menu, no logo button.
Open sliding door design, more convenient to take and put the sample.
It is characterized by small volume, low noise, even distribution of temperature field and oscillation of eight samples at a time.
The work process is fully automated, which reduces the labor intensity of the staff and improves the work efficiency. It is an ideal replacement product for the laboratory
Display: large-screen LCD (240*128);
Temperature range: room temperature ~100℃;
Temperature control accuracy: room temperature ~50℃±0.2℃;
0.3 ℃ to 50 ℃ ~ 100 ℃ +;
Oscillation frequency: 275±3 times/min;
Oscillation amplitude: 35mm;
Formula 1: after a constant temperature of 50℃ for 10 minutes, oscillate for 20 minutes, and rest for 10 minutes,
Maintain a constant temperature of 50℃;
Formula 2: after a constant temperature of 50℃ for 10 minutes, oscillate for 5 minutes and maintain a constant temperature of 75℃;
Self-definition: set the oscillation and rest time arbitrarily within 20~99 minutes;
Arbitrarily set temperature within 20~99℃;
Sample size: 8 100ml syringes;
4 250ml triangular flasks;
Power source: ac 220V±22V;
Contact Person: Weisufen