Anti-fuel spontaneous combustion point tester simple operating procedures
SH706B Automatic spontaneous combustion point tester is designed and manufactured according to the power industry standard OF the People's Republic of China DL/T 706-2017. The instrument adopts advanced microcomputer technology and color display with touch screen, adopts advanced temperature control mode, and automatically completes the determination of anti-fuel spontaneous combustion point. It has automatic constant temperature countdown, automatic sampling, automatic detection of spontaneous combustion point and automatic printing function of results
Expected flash point: Click the data area to automatically pop up the keyboard, enter the expected flash point value of the sample and click OK. set
Set the range to 100-650℃.
Sample number: Click the data area to automatically pop up the keyboard, enter the sample number, click OK. The value ranges from 0 to 250.
Automatic printing: set this parameter to 1. The test results will not be automatically printed after the test is completed. Set to 2 when the detection is finished automatically print
Current test results.
Constant temperature time: Click the data area to automatically pop up the keyboard, input the unit of time required for constant temperature is second, click
Ok then. The value ranges from 0 to 3600 seconds.
Timing time: click the data area to automatically pop up the keyboard, input the injection from the beginning to the burning phenomenon, etc
The unit of time is seconds. Click OK. The value ranges from 0 to 600 seconds
Automatic temperature increase: click on the data area to automatically pop up the keyboard, enter the expected temperature if no ignition point is detected
To increase the degree, click OK. The value ranges from 0 to 99.
Start setting: When it is found that the temperature control precision is not ideal, first set the expected flash point, click start setting, the instrument will automatically control the temperature setting, about 30 minutes, after the end of setting the instrument will automatically enter the temperature control state, observe the temperature control state is good, click stop setting, the setting is finished.
Stop setting: you can click at any time in the setting process and the setting will end.
All parameters set on this screen can be saved after power-off. You do not need to reset the parameters after the system restarts.