Anti-fuel spontaneous combustion point tester
Automatic spontaneous combustion point tester SH706B is designed and manufactured according to the requirements of DL/T 706-2017 electric power industry standard of the People's Republic of China. The instrument adopts advanced microcomputer technology and color display with touch screen, adopts advanced temperature control mode, and automatically completes the determination of anti-fuel spontaneous combustion point. It has automatic constant temperature countdown, automatic injection, automatic detection of spontaneous combustion point and automatic printing function
Expected ignition point: Click the data area to automatically pop up the keyboard, input the expected ignition point value of the sample and click OK. set
The range is 100-650℃.
Sample number: Click the data area to automatically pop up the keyboard, input the sample number, click OK. The value ranges from 0 to 250.
Automatic printing: Set to 1. The test result will not be printed automatically after the test is finished. Set to 2 to print automatically when detection ends
Current test results.
Constant temperature time: click the data area to automatically pop up the keyboard, enter the time required for constant temperature in seconds, and click
Ok then. The value ranges from 0 to 3600 seconds.
Timing time: click the data area to automatically pop up the keyboard, input the start of injection to the combustion phenomenon, etc
The unit of wait time is second. Click OK. The setting range is 0 to 600 seconds
Automatic temperature increase: Click the data area to automatically pop up the keyboard and input the expected temperature when no ignition point is detected
Click OK to increase the degree. The setting range is 0-99 degrees.
Start setting: when the temperature control accuracy is not ideal, first set the expected ignition point, click "Start setting", the instrument will automatically control the temperature setting, which takes about 30 minutes. After the setting, the instrument will automatically enter the temperature control state. When the temperature control state is good, click "stop setting" and the setting will end.
Stop setting: You can click at any time during the setting process, and the setting will end.
All parameters set on this screen have the power failure saving function, and do not need to be reset after restart.