Test method for flash point of coatings
According to the national standard GB 3186-82 "sampling of coating products", take out representative coating samples from the coating products to be tested.
Before the test, the sample should be placed in a closed container, the free volume of the container should not exceed 10% of the total volume. Samples should not be stored in plastic (e.g. polyethylene, polypropylene, etc.) bottles.
The flash point detection of paint is determined by GB/T5208 flash point measurement fast balance cup method to determine, SH105D balance method of low temperature closed flash point instrument developed and produced by Shandong Shengtai Instrument Co., Ltd. is designed and manufactured in accordance with the requirements of the People's Republic of China standard GB/T5208 "flash point measurement fast balance closed cup method". The results can be stored on a microcomputer, or the data can be stored on a USB flash drive. The instrument also meets the requirements of ISO 1523 and ISO 3679 standards.
This instrument is especially suitable for closed cup flash point in -30℃ ~ +100℃ range of all kinds of paint, paint, adhesives, solvents, petroleum and related products closed cup flash point test.