|Name:||Kangs Carbon Residue Tester||Standard:||GB/ T268 Standard|
|Round Iron Cover::||Lower High (50 ~ 53), Middle Cone High (25±2)||Model:||SD268|
|Origin:||CHINA||Inner Iron Crucible:||Capacity (75±5)ml, Black Plated On Surface|
lab test equipment,
oil analysis lab equipment
SD268 kangs carbon residue tester is designed and manufactured according to the requirements specified in GB/T 268 standard of the People's Republic of China "determination of carbon residue of petroleum products (kangs method)". It is applicable to the determination of carbon residue of petroleum products after evaporation and pyrolysis according to the method specified in this standard, so as to provide an indicator of the relative coking tendency of petroleum products.
I. main technical features
1. The design of this instrument fully conforms to the requirements of GB/ T268 standard.
2, the design is simple, complete, as long as the user is equipped with household gas (or other civil gas) can do the test.
Main technical parameters and indicators
1. Meet the requirements of GB/T268;
2. Inner iron crucible: capacity (75±5)ml, black plated on surface;
3. External iron crucible: capacity (190±10)ml, covered and coated with black;
4, tripod: iron, nickel chrome plated surface polishing, high (250 + 10) ㎜, ring diameter Ф ㎜ (130 + 5);
5, covering the flame body: the hole in the top diameter Ф ㎜ (90 + 2), the mouth diameter Ф ㎜ (82 + 2);
6, round iron cover: lower high (50 ~ 53), middle cone high (25±2),