Main performance characteristics and matters needing attention of closed flash point tester
When the test sample is used in a closed environment, the closed flash point value of petroleum products is detected. To touch screen instead of keyboard operation, using foreign advanced technology, LCD large screen full Chinese display man-machine dialogue interface, full screen touch keys prompt input, convenient and fast, open, fuzzy control integrated software, modular structure.
The closed flash point tester is designed and manufactured in accordance with the national standard and can accurately measure the closed flash point of petroleum products in strict accordance with the standard description. The closed flash point tester for petroleum products uses a closed cup to measure the closed flash point of petroleum products. Description of performance: PLD-261A closed flash point tester for petroleum products is mainly composed of test oil cup, heating plate, heating furnace, test primer and so on. The test oil cup is a standard oil cup. The heating plate is made of chrome plated steel plate. The solid-state voltage regulator is used to adjust the heating rate of the heating furnace, which is highly integrated, easy to operate, easy to install and maintain, and the output voltage is displayed directly through the voltmeter, intuitive and durable. The closed flash point tester for petroleum products has the advantages of beautiful design, reasonable structure design, convenient operation and accurate measurement.
Performance characteristics: The closed flash point tester is mainly composed of electric heating system, motor stirring device, standard oil cup, primer, electric automatic ignition mechanism and other parts. Electric stirring device adopts hard and soft shaft matching mode, stirring action is flexible, and stirring is very uniform. Electric ignition structure adopts remote control box, as long as press the switch, can automatically complete ignition, oil cup automatically open cover, close cover and other standard actions. Opening and closing time. Meet the standard description, reduce human error. Temperature control mode adopts solid state voltage regulator and potentiometer, so that the heating power can be continuously adjusted and digital display voltmeter indication, temperature regulation is intuitive and convenient.
Precautions for closed flash point tester:
1, The instrument should be used in a non-corrosive environment. When changing the sample, the oil cup must be cleaned.
2, The flash point detection ring should be kept clean, if there is oil, can be dried with filter paper to avoid affecting the detection sensitivity.
3, When the instrument is not in use, it should be placed in the temperature of 10 ~ 40℃, relative humidity below 80%, the air does not contain corrosive gases and harmful substances in the environment.